Here is a video I produced for Adult Retrieval Victoria on Hamilton’s proprietary intelligent ventilation mode, ASV, using the Hamilton T1 Ventilator, but it’s applicable to our Hamilton C1.
Please do not distribute, as these videos are AV property.
ASV takes into account the patient’s respiratory mechanics, which are measured breath-by-breath by the proximal flow sensor. ASV ensures optimal ventilation for each patient during passive ventilation but also has automatic adjustment in spontaneously breathing and weaning patients.
In passive patients, ASV is a volume-targeted pressure controlled mode with automatic adjustment of inspiratory pressure, respiratory rate, and inspiratory/expiratory time ratio. Maximum tidal volume is controlled by setting a maximum inspiratory pressure. Expiratory time is determined according to the expiratory time constant in order to prevent dynamic hyperinflation.
Watch the following video, together with the Initial Setup (Page 2) & Troubleshooting Guides below:Initial ventilator settings, Western, Draft, Feb 2020
Ventilator problems algorithm, Hamilton, Version 5, May 2018